Phone: +99890 9318616
Due to the compressing fluid flow formed elements of urine always be the focus of the lens of the microscope. The particles pass urine microscope lens and CCD cameras, one after the other, eliminating overlap. In addition, the flow of urine is diffusion, which impedes aggregation visible components.
High-speed imaging technology.
Each microscope image is flashed from the light source (40 times per second), all passing of urine formed elements fixed camera. CCD camera takes 820 pictures for each sample over time.
Identification technology formed elements of urine.
Each component of the visible image is outlined. The software immediately produces a classification of formed elements of urine from the formula, size, texture, contrast and other parameters.
Maximum relative deviation of repeatability ≤ ± 8.0%
The accuracy of the test tasks CV ~ 10%
Specificity (automatic identification)> 80%
Specificity (artificial aids) ~ 95%
Normal working condition:
Power supply: AC 220 V ± 22 V, 50/60 Hz
Operating temperature 10C ~ 30 C
Relative humidity 20% -80%
Atmospheric pressure 860 hPa ~ 1060 hPa